Skip to Content
More Contact Information
Sign Up Now
Home > CRYOGLOBULIN SCREEN
Two full 10 mL red top vacutainers warmed to body temperature, no preservatives, no serum separators. Serum from a plain red top is the only acceptable sample type. Gel barrier tubes are not acceptable.
Allow blood to clot completely at 37°C for 1 hour. Centrifuge warm (room temp is ok if 37°C is not available) to thoroughly remove cells, Then immediately transfer cell-free serum to clean tube.
Preferred volume: 10mL serum
Minimum volume: 10 mL serum
Send serum Priority Overnight via FedEx and in a well insulated container at ambient temperature.
It is CRITICAL that the specimen be kept warm while the blood is clotting. The cryoprecipitate may be lost in the clot if the blood becomes chilled.
<0.2% precipitate from original serum volume
Contact Client Services
Advanced Diagnostic Laboratories
1400 Jackson St
Denver, CO 80206
© 2017 National Jewish Health Policies & Guidelines